- LDL sensors: Conductivity – a new approach
- Installation guidelines
LDL sensors: Installation guidelines
For proper operation, some guidelines must be followed.
Preferable mounting is just before or in a vertical pipe with flow going up to ensure a full pipe. When installed in a horizontal pipe, install the sensor at a maximum of 45° from horizontal to avoid the influence of air bubbles or deposits.
Mounting in plastic pipes
The design of the LDL100 requires both electrodes to be in contact with the medium. Therefore, a metal fitting such as E43313 (G1/2 BSPP x ¾” NPT) is recommended. The LDL200 is not affected due to its design.
The measurement channel of the LDL200 must be installed parallel to the direction of flow. An alignment mark is laser etched on the body of the sensor indicating the correct orientation in the pipe.